Hi`Res-C Series Handling Instructions
Instructions for Handling High Resolution ProbesDANGER:
In order to avoid irritations and disappointments prior to using these high-end AFM tips, please observe the following instructions:
ATTENTION: ESD HAZARD!
MikroMasch High Resolution Silicon Probes are packaged in special vacuum-sealed ESD shielding bags. While the probes remain in the closed ESD protective bag, there is no risk of ESD damage.
Before opening the ESD bag in an ESD-protected area please ensure that you are grounded (e.g. with a static dissipative wrist strap or ESD-protective footwear) and the gel box is on ground potential (e.g. by using a grounded pad).
Gel boxBefore opening the gel box please ensure that the delivered probes are consistent with your order and there is no sign of transportation damage (e.g. loose AFM probes).
In case one or both of the above points are not satisfied, please contact your supplier immediately.
Probe handlingTo pick up and handle probes, we recommend using pointed Teflon tweezers.
Using the recommended tweezers helps avoid damage to the holder chip edges and ensures that the chip remains free of particles.
Proper AFM System SettingsPlease optimise the settings of your AFM system in order to ensure a smooth and gentle approach of the tip to the sample.
Tips with ultra-small radii of curvature are very fragile. ANY hard contact between the tips and ANY solid surface outside of the AFM will break the tips. Please be very careful when handling and storing AFM probes.